Nanostructure analysis using spatially modulated illumination microscopy.
Cells, Microscopy-Fluorescence, Nanostructures
see Reprint Collection (a pdf is available)
Nat Protoc 2007 2(10):2640-6.
We describe the usage of the spatially modulated illumination (SMI) microscope to estimate the sizes (and/or positions) of fluorescently labeled cellular nanostructures, including a brief introduction to the instrument and its handling. The principle setup of the SMI microscope will be introduced to explain the measures necessary for a successful nanostructure analysis, before the steps for sample preparation, data acquisition and evaluation are given. The protocol starts with cells already attached to the cover glass. The protocol and duration outlined here are typical for fixed specimens; however, considerably faster data acquisition and in vivo measurements are possible.
Baddeley, D; Batram, C; Weiland, Y; Cremer, C; and Birk, U J., "Nanostructure analysis using spatially modulated illumination microscopy." (2007). Faculty Research 2000 - 2009. 1664.